New tackiness tester
Jan 15 2020 Read 164 Times
Koehler Instrument Company, Inc. recently launched their newly minted Tackiness Tester, K95200. It has been created to provide an objective and qualitative test for tackiness. The tester adeptly quantifies the cohesive and adhesive tactile properties of greases by measuring the sample in Newton’s and replaces the old finger stick test method with a numerical and graphical result. Greases are often found in machinery and automotive components to guard against frictional and wear losses. This equates to interaction properties like performance, most saliently when it comes to bearing aspects of grease such as thickener network, wetting agents and additives.
The measurement system consists of a dual set of parallel discs which are precisely aligned. The lower disc is connected to a high precision force measurement system and the upper disc to an electro mechanical cylinder system. All components are adjusted together to make certain that the outcome is as dependable as possible. The automatic self-calibration process, before the test procedure begins, offers the maximum level of confidence. The intuitive software interacts with the user via the color-touch panel as simply as possible. A visual, guided step-by-step instruction facilitates an instantaneous start. An optional heating and cooling upgrade is available which allows grease tackiness tests at various temperatures up to a maximum of 80°C (175°F).
The specifications include the following: Accuracy of <0.1 N; Max. Test Force of 100 N; Electromechanical Load System; a sample test size of 5 g; test time of approximately 120 s; Test Speed is standard 250 mm/s; PLC is Siemens System + HMI; Ethernet Interface; Temperature Range (Optional) is ambient – 80°C (50 – 175°F); Voltage of 110-230 V/ 50 – 60 Hz; and Power: 150 W max.
The Koehler K95200 is extremely user-friendly by utilising these five easy-to-follow steps: Make sure that the discs are clean and dry; select a new test sequence on the touch screen; load the lower disc with the test sample (approximately 5 g of grease is sufficient); press the start button; view result displayed on the screen.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Fuel For Thought - Successful JASIS sees Tanaka announce appointment of new general manager - USA appointments underpin Signal Group growth plan - Leading European sensor tr...
View all digital editions
Mar 01 2020 Chicago, IK, USA
Mar 04 2020 Beijing, China
Mar 09 2020 Memphis, TN, USA
Mar 09 2020 Muscat, Oman
Mar 10 2020 Rotterdam, The Netherlands