Measurement of Manganese in Gasoline by the Rigaku NEX QC+ EDXRF Analyser
Jan 17 2018 Read 658 Times
Methylcyclopentadienyl manganese tricarbonyl (called MMT or MCMT) is an anti-knock agent added to motor gasoline and Avgas (aviation gasoline) to boost octane rating, replacing tetraethyl lead (TEL) in many regions of the world. In motor gasoline the Mn content is typically between 50-500 mg/kg, and can be as high as 3000 mg/kg (approximately 3 g/L) in Avgas. Reliably characterising the Mn content of gasoline ensure optimum engine performance based on the engine’s compression ratio and other geometrical and mechanical operating conditions. To meet the needs of the industry, Rigaku offers NEX QC+, a simple and versatile benchtop EDXRF analyser for the analysis of manganese in gasoline.
The method described here demonstrates the measurement of manganese (Mn) in gasoline by the empirical approach using the Rigaku NEX QC+ high-resolution benchtop EDXRF analyser.
The Rigaku NEX QC+ spectrometer is designed for demanding applications, or for situations where analysis time or sample throughput is critical. A silicon drift detector (SDD) affords extremely high count rate capability with excellent spectral resolution, enabling NEX QC+ to deliver the highest precision analytical results in the shortest possible measurement times.
Specifically designed for routine quality control applications, the new Rigaku NEX QC+ features an intuitive "icon-driven" touch screen interface for easy operation and a built-in printer for convenience.
Ensure each sample is homogeneous and stable. Shake sample gently and allow any bubbles to settle. Fill a 32mm XRF sample cup with 4g of sample to ensure consistent sample depth. Chemplex Etnom™ film (3um) is used, which does not degrade quickly with gasoline or oxygenated gasoline. The sample cup lid must be vented with a small hole to ensure vapor pressure does not build. Due to high evaporation rate of gasoline, it is recommended to make the measurement immediately after filling the cup. [Note: Use of internal standard is not required.]
Empirical calibration is made using commercially available certified gasoline calibration standards containing the Mn additive. A typical calibration is demonstrated here for the range 25-500 mg/kg Mn. (Please see image A)
Instrument repeatability (precision) is determined by ten repeat analyses of each sample in static position using a 100 sec analysis time per measurement. (Please see image B)
The Lower Limit of Detection (LLD) is defined as three times the standard deviation of ten repeat analyses of a blank gasoline sample containing no Mn. The following typical LLDs are reported here using 100 and 300 sec measurement times. (Please see image C)
The results shown here indicate the Rigaku NEX QC+ EDXRF analyser can be used to reliably measure Mn in gasoline and Avgas without the need to use an internal standard.
EDXRF / XRF for Petroleum & Petrochem applications from Rigaku please click here.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Fuel for Thought - New Partnership for the US Distributio - Element opens new oil and gas laboratory in Singapore - Metrohm Acquires B&W Tek Analytical Instrumentation...
View all digital editions
Oct 20 2018 Houston, USA
Oct 21 2018 Houston, Tx, USA
Oct 22 2018 Amsterdam, Netherlands
Oct 23 2018 Kiev, Ukraine
Oct 24 2018 Toronto, Canada