Measurement and Testing
EDXRF Analysis of Additive Elements in Lubricating Oils by ASTM D7751
May 16 2018 Read 417 Times
Measurement of additive elements in lubricating oils by ASTM D7751 is demonstrated.
Controlling additive elements when blending new, fresh lubricating oils is vital to ensure correct lubrication properties and minimising production waste and product rejection. Energy dispersive X-ray fluorescence (EDXRF) is a fast and simple analytical technique for use by non-technical operators, yet powerful enough for expert use in the research & development of new lube oil formulations. To meet the industry needs in production and research, Rigaku NEX CG Cartesian geometry EDXRF spectrometer offers compliance to ASTM D7751 - Standard Test Method for Additive Elements in Lubricating Oils by EDXRF Analysis.
- Model: Rigaku NEX CG
- X-ray tube: 50 W Pd-anode
- Detector: High performance SDD
- Sample Type: Fresh Lube Oil
- Film: Polypropylene, 4 µ m
- Environment: Helium Purge
- Standard: 15-position Sample Tray (32 mm)
- Analysis Time: 700 sec
Ensure each sample is homogeneous and stable. Simply fill a 32 mm XRF sample cup with 4.0 g of sample.
Empirical calibrations were built using a suite of 23 commercially available calibration standards. The suite of calibration standards must be representative of the lube oil formulation to be analysed. Elements in the lube oil should vary evenly over each concentration range of interest, and the elements in the oil should vary independently of each other. Empirical alpha corrections are then employed to automatically compensate for variations in X-ray absorption and enhancement effects within the sample due to the independent variations in element concentration.
The empirical method was used to determine estimated detection limits. In the empirical method, ten repeat analyses of a blank lube oil (or mineral oil) sample are taken with the sample in static position and the standard deviation is determined. The Lower Limit of Detection (LLD) is then defined as three times the standard deviation. The following typical LLDs are reported here for the lube oil formulation shown above using a total analysis time of 700 sec.
Two representative samples from the calibration suite were chosen to demonstrate typical instrument repeatability (precision). Ten repeat analyses of the sample were performed with the sample in static position using a count time of 100 sec per analysis condition.
The Rigaku NEX CG spectrometer combines indirect excitation with secondary targets, polarization targets and a high-performance SDD detector to yield the optimum performance for elemental analysis of lube oils. Measurement times can be adjusted based on the formulation being measured to optimise various groups of elements. Rigaku RPF-SQX fundamental parameters can also be used in association with matching libraries to encompass families of lubricants in a single method and for use with R&D testing new formulations, making the NEX CG analyser an excellent tool for process and quality control in the manufacturing of lube oil formulations and blends, as well as R&D and other uses.
Click here to request a brochure today.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Fuel for Thought - New Partnership for the US Distributio - Element opens new oil and gas laboratory in Singapore - Metrohm Acquires B&W Tek Analytical Instrumentation...
View all digital editions
Sep 24 2018 Dallas, Tx, USA
Sep 24 2018 Rio De Janeiro, Brazil
Sep 25 2018 Kielce, Poland
Sep 25 2018 Calgary, Alberta, Canada
Sep 25 2018 Kuala Lumpur, Malaysia