Flow Level Pressure
Understanding installation problems with ultrasonic clamp on flowmeters
Feb 12 2020 Read 504 Times
Young Calibration was contracted to undertake a simple correlation study on the user installation variation when installing ultrasonic clamp on meters for fluid flow measurements. Meters were installed to the manufacturer’s guidelines by four different field technicians on a 100 mm pipeline. The study was conducted using two transit time type meters; a low-cost model single channel model and a high-performance dual channel meter. The findings highlighted a large sensitivity to user installation on the low-cost single channel model, with a flow rate error variance of 5 to 32% whilst the high-performance dual channel clamp-on meter recorded error variance of -0.5 to 3%. In the first of our application notes for ultrasonic clamp on meters it is astoundingly clear that meter quality is a significant contributor to attaining accurate flow measurement results. In future parts we will be discussing installation effects in more detail; with traverse type set-up, coupling pastes, pipe sizes, particulate/bubbles, pipe condition and how meter diagnostics can help you achieve better results.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Safety - BM 25 & BM 25 Wireless: 10 years of improved safety - Discover the enhanced PS200 Fuel For Thought - FCI announces Adam Schleyhahn as Director of Sales - New...
View all digital editions
Jul 27 2020 Virtual event
Jul 27 2020 Kintex, South Korea
Jul 28 2020 Bogota, Colombia - NEW DATES
Aug 03 2020 Lombard, IL, USA
Aug 03 2020 Moon Township, PA, USA