Best Practices for Improved Sulfur Repeatability
May 02 2019 Read 324 Times
Author: X Ray Optical Systems (XOS) on behalf of X Ray Optical Systems (XOS)
Free to read
Articles are free to download. Please login to read this article or create an account.
X-ray fluorescence (XRF) is a technique that is known for its ease-of-use by not requiring injections, sample digestion, or combustion. Within the broader scope of elemental analysis, XRF is seen as having a lower threshold for training technicians. XRF sample preparation, however, is still dependent upon its operators to properly prepare their samples and input the data error-free to ensure that the lab is reporting accurate data.
Proper sampler preparation is perhaps the most critical component to improve repeatability. Poor sample preparation and contamination when preparing samples are the primary causes for repeatability issues.
Download our whitepaper to learn about proper sample preparation techniques including cleanliness, measurement time, venting, repeat measurements, and calibration - among others.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Fuel For Thought - Successful JASIS sees Tanaka announce appointment of new general manager - USA appointments underpin Signal Group growth plan - Leading European sensor tr...
View all digital editions
Feb 18 2020 Abu Dhabi, UAE
Feb 25 2020 Erbil, Iraq
Feb 25 2020 London, UK
Feb 25 2020 Lagos, Nigeria
Feb 26 2020 Guanghzou, China